Questmed - Shoulder Endoprosthesis - Wear and Fatigue TestQuestmed GmbH offers testing for wear of shoulder endoprosthesis according ISO 14243.
We are also specialised for testing anatomical shoulder endoprosthesis and reverse shoulder endoprosthesis.
As well as we offer fatigue testing according ASTM F1829 and ASTM F2028.
Our Shoulder Wear Simulator is available with two banks of three test stations. Our AMTI simulator (ADL-K6-01) replicates the loading and full range of multi-axis motion associated with activities of daily living. The test system provides unparalleled test accuracy through its Virtual Soft Tissue control, which couples force control with a software model that accounts for the constraining influence of the tissue structure surrounding the shoulder. The AMTI simulator meets the requirements for shoulder wear testing.
Parameters of the Wear Test
The tests will be conducted load controlled so that the flexion and loads specified in motion cycle will be applied to the test samples. The axial force specified in motion cycle will be applied to the active load soak sample.
- Axial Load 200 N – 600 N (motion cycle)
- Abduction / Adduction -10 degree – +40 degree (motion cycle)
- Translation 0 N- +256 N (motion cycle)
- Elevation -1 Nm – +1 Nm (motion cycle)
- Test Frequency 1 Hz
- Test Temperature 37°C +/- 2
- Test Medium Calf serum (Protein content 20 g/l)
- Cycles 5 000 000
All parameters and setup will be confirmed by the customer.
The wear of the polyethylene components will be determined by the net mass loss Wn according ISO 14243-2. The polyethylene components will be weighted before and after the wear test as well as during the wear test overall seven times. The inspections will be determined after 500 000 cycles, 1 million cycles and every 1 million cycles.